|Place of Origin:||China|
|Minimum Order Quantity:||0-100|
|Packaging Details:||Carton / Wooden Box|
|Delivery Time:||5~8 working days|
|Payment Terms:||L/C, D/A, T/T, Western Union, MoneyGram|
|Measuring Range:||0-1000mg/L NH4-N||Response Time:||10 Sec|
|Protection Level:||IP68||Deepest Depth:||Underwater 60 Meters|
|Temperature Range:||0 ~ 50℃||Output:||Support RS-485, MODBUS Protocol|
|Size:||Φ26*177.5 Mm||Probe Cable Length:||10 M, Customizable 5 M, 15 M And 30 M Cable|
water turbidity sensor,
oil in water sensor
Hot Sell Good Quality Water Treatment Online Dissolved Oxygen Electrode Sensor DO Meter Probe
The ammonia nitrogen sensor is an online device for detecting the content of ammonia nitrogen (NH4+- N) in water, which is applicable to various water bodies, including lakes, streams, groundwater and wastewater. The sensor measures the ammonia nitrogen value through the electrode method. The NH4+electrode provides the main measurement. It measures the ammonia ion (NH4+).
Kacise introduces an online ammonia nitrogen sensor that requires no reagents, is green and non-polluting, and can be monitored online in real time. Integrated ammonium, potassium (optional), pH and reference electrodes automatically compensate for potassium (optional), pH and temperature in water. It can be directly put into installation, which is more economical, environmentally friendly and convenient than traditional ammonia nitrogen analyzer. The sensor has a self-cleaning brush that prevents microbial adhesion, resulting in longer maintenance intervals and excellent reliability. It adopts RS485 output and supports Modbus for easy integration.
Digital sensor, RS-485 output, support MODBUS
No reagents, no pollution, more economical and environmentally friendly
Automatically compensates for pH and temperature in water
Self-cleaning brush to prevent microbial adhesion and longer maintenance intervals
|Measuring range||0-1000mg/L NH4-N|
|Response time||10 sec|
|Deepest depth||Underwater 60 meters||underwater 10 meters|
|temperature range||0 ~ 50℃|
|Output||Support RS-485, MODBUS protocol|
|Power supply||DC 5~12V, current <50mA|
|Probe cable length||10 m, customizable 5 m, 15 m and 30 m cable|
1) Sensor outer surface: Clean the outer surface of the sensor with tap water. If there is still debris, wipe it with a soft, damp cloth.
2) Check the cable of the sensor: the cable should not be tightened during normal operation, otherwise the cable inside the cable will be broken, causing the sensor to not work properly.
3) Sensor storage: When the sensor is not in use, the black plastic cap should be fastened to check whether the sponge inside is wet. If it is not wet enough, please add 3mol/L potassium chloride solution and let the electrode be placed with potassium chloride. Preserved in the solution.
The Matters Needing Attention
The probe contains sensitive optical and electronic components. Make sure the probe is not subject to severe mechanical shock. Damage to the sensor due to impact or man-made will not be covered by the warranty. The sensor inlet water is not covered by the replacement of the pH electrode.
Contact Person: Ms. Evelyn Wang
Tel: +86 17719566736
Address: i City, No11, TangYan South road, Yanta District, Xi'an,Shaanxi,China.
Factory Address:i City, No11, TangYan South road, Yanta District, Xi'an,Shaanxi,China.